PAK – CHANNEL OVERVIEW
The purpose of this PAK Tip is to present the “Channel Overview” functionality which provides access
to the information contained in a measurement file and quick display of 2D representable analyses.
With the PAK feature “Channel Overview”, all information contained in a measurement file is accessible : channels used / label of measurement points and direction / quantities...
Furthermore you can quickly display analyzes representable in 2D (as a function of time or of a Tracking parameter) of all or part of these channels.
This functionality is used on the one hand in the Data Viewing (general information) and on the other hand in the Graphic Definition (display of 2D analyzes).
1. DATA VIEWING
In the "Data Viewing", the "Channel Overview" feature allows access to information about the measurement channels :
- channel number,
- label position,
- presence of overload,
2. GRAPHIC DEFINITION
Using the "Channel Overview" feature into the Graphic Definition, you can display the results of 2D analyses as a function of time or of a Tracking channel in a 3D diagram.
For example :
- Sum Level,
- Band Pass Mag,
- Order Mag,
- Octave channel,